Product Description: V-MAX(HIFU) helps to improve wrinkle and skin elasticity and to form collagen. HIFU thermal energy at 65℃ is delivered to SMAS layer (about 3.0mm~4.5mm) in the face skin and it contracts skin muscle layer. Moreover, V-MAX enables to do obesity treatment by breaking down fat layer which is in between 10mm~20mm depth in the body skin.
Features: As it is equipped with the Probe (3.0mm/4.5mm) which is the most widely used in HIFU equipment, it can be used for both facial and body treatment. By adopting a self-cooling method, not a cartridge-changing way which is widely known in the market in normal HIFU devices , V-MAX can be used for a long time without replacing a consumable cartridge. (Durable handles) Probe-rubbing methods enable easy and particular region operation. Easy control though user friendly interface on 7 inch touch LCD
Advantages: Because V-MAX HIFU focuses energy shortly and intensively on the target area while rubbing probe, it makes less pain than other HIFU brands`. Various shot intensity, shot time and shot interval can be adjusted by user`s purpose. As applying rubbing operation, reducing shot and interval time, operation time can be shorter than normal HIFU operation. These shorter operation time enables to do more operations and it helps to get good results rapidly. V-MAX doesn`t need maintenance cost which mostly made by cartridge changing. It reduces medical expense and increases competitiveness. And it helps to do additional treatment without big burden. By adopting a probe-rubbing method which is not widely used in HIFU equipment, it is possible to perform detailed operation. Ex) near cheek-bone Water cooling system enables to give stable operation, though a long time working
It has the effect of facial , product lead-in, promoting the regeneration of fascia layer and Collagen)(Before using the machine , please read this instruction and doing the treatment according to the guidance seriously
Product dimensionï¼?2*39.6*173mm Power supply voltageï¼?00~240 Powerï¼?0W Working voltage：DC24v Standard：GB4706.1-2005 Operating handle : 1
Operation procedure: 1.Firsly , you need to clean the treatment area(If it has make up , you need to remove make up then using cleanser to clean the area) 2. Connect the power line , Press the round button to turn on the machine(There has a large round button on each side of the button, press any one of them can turn on the machine , press it when the light turn to light , the machine is on , if you need to turn off the machine , press the button to 1.5 seconds) 3.The depths and energy of the machine can be adjusted when the machine turns on, We suggest 4.5mm to operating on face. 4.The energy of the Mechanical wave and RF have 3 levels: Low , Medium , High. The round button on the handle can adjust the power of the RF. The reverse button can adjust the power of the Mechanical wave. The round button below are the Operation button/ pause button. Press the operation button , will have 30 seconds automatic output, During the output , all the button are unavailable ,and the buzzer will sound once every second. If it doesn’t sound , it means it has finished the 30 seconds output. If pause needed during the period of output, press the pause button. The square push button can adjust the depth. There are 3.0mm/4.5mm respectively , the red line is corresponds to the depth 5. Testing the energy intensity: First pull up from the right lower jaw to the top of mouth in a stable speed, If you feel nothing , you can adjust the energy of Mechanical wave and RF, It is normal that you feel heat , tingling or sore. When operating to the wing of the nose , the energy need to be lower.(Different treatment area can bear different energy , You need to adjust the energy according to the customer's tolerance ) Normal Skin: For the first time using Mechanical wave and RF , Needs to start from the lowest intensity, And can adjust the intensity according to the maximum tolerance during the treatment. Red bloody skin: For the first use , we suggest only turn on Mechanical wave. After using 3-4 times , you can turn on the RF with the Mechanical wave at the same time, RF energy start from the lowest intensity, and adjust the power of Mechanical wave according to your maximum tolerance. Sensitive cuticle slants thin skin: For the fist use, Lowest power of Mechanical wave and RF are suggested, After using 2-3 times , Can adjust the energy to your maximum tolerance according to personal skin situation. 6.During the treatment ,it is necessary to apply the medium on the treatment area, the cartridge needs to closed to the skin , with regular speed and lifting method, the medium needs to be supplied when it is absorbed or dried. 7. Usually it takes 30-45 minutes for one operation of the whole face.(The operating time changes according to the size of face. ) 8.Cleaning the cartridge with wet towel is needed after the operation. 9.After the operation , do clean the face with warm water or Clear water, apply one moisturizing mask right after the operation. about 10-15 minutes is ok , if the moisture has been absorbed or dried , you need to supply it (hot water not suggested) Noted: Other parts needs operation , please consult the professional beauty guidance.